Re: Defining the type of Applied Part - Metal Probe (Applied Part) employs Water Cool
Yes, absolutely, the equipment should be Type B.
For reference, dialysis equipment is normally rated Type B. It would be good for dialysis to be BF or CF, but it's not really practical as the blood circuit is electrically connected to the dialysate circuit which has various electrical sensors, solenoid valves, heaters and in turn connects external water supplies and the like. So, sometimes Type B is all that can be done.