AMS 2750F Section 22.214.171.124 g states that the "Error" shall be documented on the calibration report. We have always had the Correction Factor reported and never had it questioned by Nadcap until this year. Before having my calibration source complete redo the report to include the Error, I wanted to see if there were other options. Has anyone run into this in AMS 2750 journey? We only use the CF so I'm trying to eliminate confusion for the operators by adding another value on the cert. Is a comment on the cert saying something to the affect "Thermocouple readings provided in "correction factor" format. The Error is the opposite value of the Correction Factor."