Calculating Coefficients for an RTD (Resistance Temperature Detector) Probe

M

MetroBioCen

Hello everybody We sent an RTD to be calibrated. We used this RTD to calibrate our thermometers. Althought the Calibration Certificate is Ok, the coeficients for the RTD are not present. We need those corfficients to be set on the Hart 1502A digital thermometer which uses the RTD, in order to measure temperature. How can I calculate those coefficients. Values of resistance vs temperature are as follows: 84,070Ω -39,965?C 100,1346Ω -0,008?C 139,4873Ω 100,068?C 196,2719Ω 250,119?C 318,3385Ω 599,647?C Thank you very much in advance MetroBioCen

dgriffith

Quite Involved in Discussions
Re: Calculating Coefficients for an RTD Probe

I can't imagine sending a ITS-90 PRT thermometer to a reputable provider and not getting coefficients.
But you're calling it an RTD. Please explain what it is you really sent for cal. Was it a PRT, or an RTD, such as a PT100/IEC-751/DIN-43760?
If RTD, then the coefficiants for those are R0, Alpha, Beta, and Delta:R0=100.1346; Alpha=0.00385; Beta=1.507; Delta=0.111, according to the 1502 manual.
If you have a triple point of water cell (0.01 ?C) you can determine your own R0 and be close enough. If you need it more precise at 0 ?C and can make an ice bath very well, then do that.

Otherwise, if you sent a PRT, then: you got hosed; you didn't ask for the right calibration; you need to look on page two . . .

Last edited:
M

MetroBioCen

Re: Calculating Coefficients for an RTD Probe

Hi, thank for your answer. The device is in fact a PRT with the following data: Manufacturer: Fluke; Model: 5626. It was calibrated at TESTO Industrial Services GmbH, in Germany. I really ignore why they did not send the coefficients.

dgriffith

Quite Involved in Discussions
Re: Calculating Coefficients for an RTD Probe

We have exactly that PRT, which is in fact a secondary standard probe. It is constructed internally just as a SPRT would be, and is very fragile despite the SS sheath. Hart's admonishment is "if you can hear it when you put it down, you've put it down too hard".
Indeed you should have received the derived coefficients (A7,B7,C7, a4,b4) and a 1? conversion table.
Hart is also in England, if it helps.
In any case, that probe needs to go to a specialized thermometry provider, in my opinion.

Call the provider and ask questions.

T Formulas for Calculating Coefficients for RTD / PRTD Calibration According to ITS-90 General Measurement Device and Calibration Topics 5
Calculating Risk Estimation ISO 14971 - Medical Device Risk Management 29
Calculating Ppk for Design Verification - Variable Sampling Design and Development of Products and Processes 15
Help needed in choosing the method of calculating the minimum sample size Internal Auditing 12
True position, calculating theoretical distances General Measurement Device and Calibration Topics 0
Lean: Conducting Capacity Study, calculating Cycle times on laser cutting machines Lean in Manufacturing and Service Industries 2
Calculating Carbon Footprint in the Organization Miscellaneous Environmental Standards and EMS Related Discussions 2
Calculating your KPIs ISO 13485:2016 - Medical Device Quality Management Systems 7
Calculating Heat Dissipation Manufacturing and Related Processes 1
Calculating Defect Rates ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 4
Formula for Calculating NoGo Major Diameter for UN gages Inspection, Prints (Drawings), Testing, Sampling and Related Topics 4
Calculating a weight for a machine for the CE label CE Marking (Conformité Européene) / CB Scheme 0
Remote Support - Calculating Number of Employees IATF 16949 - Automotive Quality Systems Standard 2
Calculating Tolerance of a Panel Meter with a 2 degrees of an Arc General Measurement Device and Calibration Topics 0
Calculating LCL, UCL, Cp, and Cpk in an Excel Spreadsheet Statistical Analysis Tools, Techniques and SPC 3
Product Development - When to start calculating Process Capability Capability, Accuracy and Stability - Processes, Machines, etc. 3
Calculating Reliability for Subsystems in Series Reliability Analysis - Predictions, Testing and Standards 15
Calculating (3rd Party) Audit Days for Company with Seasonal Employees General Auditing Discussions 3
J Calculating part variation from historic data for GRR study Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
P Purpose of calculating Ta, Tb & R Square in Linearity Study Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
P Purpose of calculating Uncertainty value in calibration study Measurement Uncertainty (MU) 5
M Calculating MTBF for plug-in PCI cards Reliability Analysis - Predictions, Testing and Standards 2
J Calculating impact force, expressed in lbs, of a 5 lb weight dropped 48 inches Inspection, Prints (Drawings), Testing, Sampling and Related Topics 1
M Calculating Cpk when sample size equals to 1 Capability, Accuracy and Stability - Processes, Machines, etc. 12
D Calculating Cpk on Non-Normal Data Distribution Capability, Accuracy and Stability - Processes, Machines, etc. 10
K Calculating Capability of a process when data is skewed in the upper range Capability, Accuracy and Stability - Processes, Machines, etc. 4
A Calculating Accuracy for a Multimeter General Measurement Device and Calibration Topics 1
T Calculating Plating/Coating Weight for IMDS RoHS, REACH, ELV, IMDS and Restricted Substances 3
D Calculating Lower and Upper 3 Sigma Control Limits Statistical Analysis Tools, Techniques and SPC 2
M Calculating Adequate Receiving Inspection Sample Size Statistical Analysis Tools, Techniques and SPC 2
M Calculating Capability of Delivery Performance Capability, Accuracy and Stability - Processes, Machines, etc. 5
K Advice on Calculating Control Chart Control Limits Statistical Analysis Tools, Techniques and SPC 13
B Calculating Combined DPMO and Sigma Level for Two or More Different Work Areas Six Sigma 3
G Calculating the Fraction Defective Confidence Interval for a Lot Statistical Analysis Tools, Techniques and SPC 2
A Calculating Combined Measurement Uncertainty - VDA 5 Measurement Uncertainty (MU) 1
B Calculating Precision to Tolerance Ratio Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 9
D DPPM - Calculating the Defect Rate of a Software Test Statistical Analysis Tools, Techniques and SPC 9
L FMEA - Calculating RPN = S*O*D FMEA and Control Plans 3
I Shewhart Constants vs Central Limit Theorem in calculating Control Limits Statistical Analysis Tools, Techniques and SPC 18
B Please share a template for calculating Cp Cpk Document Control Systems, Procedures, Forms and Templates 3
S Is a Stable Process (within Control Limits) required for Calculating Pp, Ppk? Capability, Accuracy and Stability - Processes, Machines, etc. 6
Calculating the IATF TS 16949 Recertification Audit Man Days - Clarification ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 6
IEC60825 - Laser Safety - Calculating Maximum Permissible Exposure Correctly Other ISO and International Standards and European Regulations 1
P Calculating Process Capability from Data of attached spreadsheet Statistical Analysis Tools, Techniques and SPC 4
I AQL Formula in Excel - Calculating Sample Size based on an AQL Chart AQL - Acceptable Quality Level 8
D Formula for Calculating Warp Measurement of Corrugated Boxes General Measurement Device and Calibration Topics 1
S Need Help Calculating and Implementing Cs - Wright's Index Statistical Analysis Tools, Techniques and SPC 11
S Calculating an Uncertainty Budget for an Optical Comparator General Measurement Device and Calibration Topics 4
D One-Two-Three Sigma Control Limits - Calculating Control Limits for X-bar Charts Statistical Analysis Tools, Techniques and SPC 3
K Calculating (Approximate) Parallelism Inspection, Prints (Drawings), Testing, Sampling and Related Topics 10