Retired Old Goat
I'm not sure what to say. If anyone else reading this is experiencing the same problem with any phone, hopefully they will report it in this thread.
|Thread starter||Similar threads||Forum||Replies||Date|
|S||Cannot create Rejects for TMV on a medical instrument with an LCD||Other Medical Device and Orthopedic Related Topics||10|
|K||What is supposed to happen when a CA cannot be closed out or the root cause...||General Auditing Discussions||1|
|Cannot determine what type of medical device this product is....||21 CFR Part 820 - US FDA Quality System Regulations (QSR)||2|
|K||Anyone who audits 8.2.2 cannot audit other clauses of the standard?||AS9100, IAQG 9100, Nadcap and related Aerospace Standards and Requirements||31|
|If NB's (notified bodies) cannot be trusted how can we?||EU Medical Device Regulations||39|