# Concentricity of two ID's (Inside Diameters)

A

#### awall

Hello, I am trying to measure concentricity between two ID's. I've attached a snippet of the print so you can see exactly how it is called out.

I have a gage with 4 probes. Two probes measure the 96 ID and two probes measure the 134 ID. You place the part on the gage and spin the part 360?. It gives you the average measurement of the two diameters. The gage is ran using gage metrics software.

Using this gage is it possible to measure concentricity as called out on the print? If so, how?

This is measured on a CMM once per shift but the customer is requiring 100% inspection on concentricity between these two ID's. I would like the data from the gage to correlate to our CMM.

#### Attachments

• Concentricity.pdf
124.2 KB · Views: 493

#### Michael_M

Trusted Information Resource
Can not really see how the functional gage you are using can be used to measure the concentrically unless the probes are on the same center-line as datum -A-. In general, concentricity cannot be checked with a functional gage.

Can you get with the customer and use process control to set concentricity. If you are using a lathe to turn both diameters and you turn them both at the same time (same operation), it would be very difficult to fail on the .05 concentricity. You may be able to allow the CMM check once per day to ful-fill the requirements.

#### Proud Liberal

##### Quite Involved in Discussions
A lot depends on how you are holding the part and picking up the data from the probes. Since concentricity requires an analysis of centerlines, it doesn't sound like your setup can account for establishing datum -A- RFS (two probe points on a diameter wouldn't work).

I'd check the Cpk of the concentricity and approach the customer with the statistical argument to reduce the sample frequency to less than 100%. Assuming this is a machining operation and both diameters are turned without un-chucking the part, the relationship between the two should be a function of the condition of the turning center (and for all practical purposes constant).

Has this dimension been a problem in the past?

A

#### awall

Thank you both for your quick replies.

I've attached a picture of the gage we have so you can get a better idea of what I have to work with. I thought since the gage would be measuring both diameters from the same centerline I could use it somehow to measure concentricity.

This part is machined on a CNC, both dimensions are cut without being removed from the chuck.

This dimension hasn't been an issue for us historically, but we haven't been able to get the customer to budge on letting us reduce the frequency of the check, the language barrier makes this extremely difficult. Currently the gage and CMM concentricity data do not correlate and this is causing a major headache for Quality and Production trying to produce good parts.

A

#### awall

Forgot to attach picture, sorry about that.

#### Attachments

• gage.jpg
101.5 KB · Views: 442

#### Golfman25

Trusted Information Resource
I don't think you'll be able to do it with that gage. Unless you can develop some mathematical formula between the two diameters and the centerline.

Probably easier to just make up a new gage to measure concentricity - and get a direct measure. Or check it while in the CNC.

Good luck.

Z

#### zancky

imho If the requirement is changed adding two maximum material principle, you can check concentricity with a stepped plug gauge

#### Stijloor

Super Moderator
To all the posters above: What GD&T Standard does apply?

The definition per ASME Y14.5 is quite different from ISO.

Virtually ALL concentricity callouts on ASME-based drawings are wrong/misapplied.

Failing concentricity Manufacturing and Related Processes 3
Parallelism, Perpendicularity and Concentricity Inspection, Prints (Drawings), Testing, Sampling and Related Topics 15
J Micro Tubing Concentricity Checking using Crosshairs Manufacturing and Related Processes 1
M Position vs. Concentricity vs. Runout - Measurement of a Tube Inspection, Prints (Drawings), Testing, Sampling and Related Topics 4
A What is the Standard Tolerance for Concentricity? Various Other Specifications, Standards, and related Requirements 7
J GD&T - Concentricity Call Out on a Drawing --> .005 TIR Inspection, Prints (Drawings), Testing, Sampling and Related Topics 15
L Countersink Concentricity Inspection, Prints (Drawings), Testing, Sampling and Related Topics 5
R Query on Concentricity Inspection Manufacturing and Related Processes 11
Concentricity Tolerance: A Unilateral Boundary Situation (Minitab) Using Minitab Software 22
J Blueprint Note: Rectangular Distribution is Permissible - Concentricity of Two Holes Capability, Accuracy and Stability - Processes, Machines, etc. 5
N Concentricity is Out of Specification Manufacturing and Related Processes 15
CMM Filter Settings - Measuring perpendicularity and concentricity Inspection, Prints (Drawings), Testing, Sampling and Related Topics 5
Is there a standard concentricity? Manufacturing and Related Processes 9
B Dimensions and Tolerance Callout - Straightness of shaft - Concentricity Manufacturing and Related Processes 1
D Cpk Studies for Concentricity - Dimension that is spec'd 0.0 - .03 Capability, Accuracy and Stability - Processes, Machines, etc. 15
N Concentricity - Down & Dirty Method? Inspection, Prints (Drawings), Testing, Sampling and Related Topics 4
S Explanation of Concentricity and Eccentricity Inspection, Prints (Drawings), Testing, Sampling and Related Topics 19
To Verify Concentricity of Flats to a Centerline General Measurement Device and Calibration Topics 9
How to Measure Concentricity Correctly Inspection, Prints (Drawings), Testing, Sampling and Related Topics 27
Outgassing - product inside a sealed housing Manufacturing and Related Processes 1
QMS inside a Non medical device firm ISO 13485:2016 - Medical Device Quality Management Systems 4
Warranty periods of the batteries inside the device EU Medical Device Regulations 5
60601-1 : Integrated Dry ECG Electrodes = 2 Patient connections inside 1 applied part? IEC 60601 - Medical Electrical Equipment Safety Standards Series 5
Measuring FIM (TIR) - Two inside diameters - Conflicting readings between inspectors Manufacturing and Related Processes 1
Inside the Bubble 2019 Other US Medical Device Regulations 8
S Tracking laptop movement inside the office IEC 27001 - Information Security Management Systems (ISMS) 7
D Use of hand cream inside gloves in the cleanroom for medical devices Occupational Health & Safety Management Standards 7
M Regulations for oxygen/air-tubing and connectors inside of ME equipment IEC 60601 - Medical Electrical Equipment Safety Standards Series 1
T Can incoming and outgoing areas be allocated inside the store? Other ISO and International Standards and European Regulations 1
Scope Ideas? Can our scope contain bullets? Example inside. ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 5
J Electronic Components inside a Wooden Case Manufacturing and Related Processes 4
B Temperature/Humidity inside a store Human Factors and Ergonomics in Engineering 7
Using STERILE symbols on outside and inside packaging EU Medical Device Regulations 1
A "sieve" inside a pot for seperating Tissues from Fluids - IVD or not? EU Medical Device Regulations 6
Customer Service/Inside Sales Will Not Leave Us (Quality Assurance) Alone! Document Control Systems, Procedures, Forms and Templates 3
K Desiccants & Humidity Control inside ocean going Shipping Containers Supplier Quality Assurance and other Supplier Issues 5
Propagation of Errors in Calculations - Thickness of a coating material inside a tube Using Minitab Software 6
Automobile 'Black Box' Data - An Inside Look World News 0
A When does a project end inside a software product? Life Cycle issue IEC 62304 - Medical Device Software Life Cycle Processes 5
K Is it a nonconformity if an auditor saw ant trails on wall inside the plant? Food Safety - ISO 22000, HACCP (21 CFR 120) 22
M IPC-A-610C/D Question Inside Other ISO and International Standards and European Regulations 2
D Flame Rating for Internal Plastic Support inside of a Fire Enclosure IEC 60601 - Medical Electrical Equipment Safety Standards Series 4
Q No Work Instruction inside Ambulance - Is it an NC? Nonconformance and Corrective Action 15
J IFU inside or outside Ortho Implant Sterile Device EU Medical Device Regulations 3
S High Voltage inside Ultrasonic Probes - Do SIP/SOP ports require isolation? IEC 60601 - Medical Electrical Equipment Safety Standards Series 2
T What is the significance of a drawing note number appearing inside a box Inspection, Prints (Drawings), Testing, Sampling and Related Topics 5
T Vernier Caliper Accuracy Issue - Inside measurements were slightly off General Measurement Device and Calibration Topics 7
Lead acid Batteries inside DG set acoustic enclosure - What Hazard and Risk and how ? Occupational Health & Safety Management Standards 5
S Looking for Aerospace Specs - see inside AS9100, IAQG, NADCAP and Aerospace related Standards and Requirements 6
C ID (Inside Diameter) Threads Shrinking due to Heat Treatment and Weld Manufacturing and Related Processes 11