Diameters measurement with 3D touch machine like a CMM

Marco Bernardi

Mark_Navigator
Dear all,

I'd like to solve and clear some doubts about diameters measurements like in the thread title:

when you measure with a CMM e.g. a diameter of a piece, usually is used the best-fit algorithm to get the best approximated diameter value and the roundness error to evaluate the circle form.
Then, but could not be useful to get the minimum circumscribed and the maximum inscribed circles, referred to the roundness error and not only the roundness error?
And more, for a coupling analysis with another part, should not be more indicated the minimum and maximum diameters got from the circle form construction? (that is different from roundness error that considers the circle center, instead of singles and separates diameters, that has the barycentre translating)

Marco Bernardi

Mark_Navigator
Up, up, thank you to everyone wants to go on with the thread.

John Predmore

Trusted Information Resource
The best-fit circle is not always the best choice, the answer depends on what information you want and what you will do with the information. If you measure the outside diameter of a shaft which will be inserted into a hole, the minimum circumscribed circle could be better to predict how the shaft will fit in the hole. I think you know this. What is your question?

Ron Rompen

Trusted Information Resource
Most CMM software that I have used in the past few years has the option of reporting Min circumscribed/Max inscribed/Average (least squares) diameters. If your CMM doesn't give you these options, then perhaps you should investigate a software upgrade.

Marco Bernardi

Mark_Navigator
The best-fit circle is not always the best choice, the answer depends on what information you want and what you will do with the information. If you measure the outside diameter of a shaft which will be inserted into a hole, the minimum circumscribed circle could be better to predict how the shaft will fit in the hole. I think you know this. What is your question?
You're right John, it depends which application is intended for that diameter.
However, what I want to clarify then is that max inscribed and minimum circumscribed diameters, could and usually have different center than the minimum and maximum diameters got to evaluate the roundness value with the least-squares actual diameter calculation, isn't it?
Another question: why is not considered with GD&T to give out the information of those min. and max. diameters got with least-squares calculation, instead of only roundness value, that could be positioned in different ways compared to the actual diameter got?

C Measurement System to measure Hole Diameters, True Positions, Surface Profiles, etc General Measurement Device and Calibration Topics 4
Measuring FIM (TIR) - Two inside diameters - Conflicting readings between inspectors Manufacturing and Related Processes 1
Dimensioning diameters Inspection, Prints (Drawings), Testing, Sampling and Related Topics 2
Laser Micrometer Suggestion (if any) for diameters that are +/-.0001 inch General Measurement Device and Calibration Topics 1
S How to check runout on Id to ID diameters Inspection, Prints (Drawings), Testing, Sampling and Related Topics 2
A Concentricity of two ID's (Inside Diameters) Inspection, Prints (Drawings), Testing, Sampling and Related Topics 7
L Inspection Equipment Accuracy - Measuring the diameters of small holes Measurement Uncertainty (MU) 5
C Measurements of Major Diameters of plug thread gages General Measurement Device and Calibration Topics 12
E Floor Gaging for Internal Thread Major Diameters Inspection, Prints (Drawings), Testing, Sampling and Related Topics 6
F SPC Data Analysis - Excel Spreadsheet I can use for Diameters and Perpendicularity Capability, Accuracy and Stability - Processes, Machines, etc. 6
Tolerance for Cpk purposes on certain diameters Capability, Accuracy and Stability - Processes, Machines, etc. 10
H Measuring Diameters in Coordinate Measuring Machine (CMM) Inspection, Prints (Drawings), Testing, Sampling and Related Topics 11
R Inspection of Diameters on a CMM (Coordinate Measuring Machine) Inspection, Prints (Drawings), Testing, Sampling and Related Topics 13
T Deviation of shaft and housing bore diameters? Manufacturing and Related Processes 3
C SPC presentation - How to convince someone to use SPC - Grind diameters Statistical Analysis Tools, Techniques and SPC 6
Screw Thread Major and Minor Diameters - Parts vs. Gages Inspection, Prints (Drawings), Testing, Sampling and Related Topics 9
R SPC for Diameters - Parts not Perfectly round so charts are useless Statistical Analysis Tools, Techniques and SPC 28
P Gauge R&R of automeasuring machine for checking the crankshaft diameters Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
J Conducting Gage R&R - 300 plug gauges - Different diameters - One for each plug gage? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 10
J Finding a Radius - How to measure diameters that are not a full diameter Registered Visitor Articles Archive 19
Building my own measurement equipment General Measurement Device and Calibration Topics 3
Test Method Validation (TMV) for all Measurement Methods in Rec/Inspection Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 0
Keyence Measurement Systems General Measurement Device and Calibration Topics 3
Measurement Equipment Revocation - Looking for a Disposal Form with Risk Assessment IATF 16949 - Automotive Quality Systems Standard 10
IVD Metrological traceability without a reference measurement procedure General Measurement Device and Calibration Topics 11
Any template/ form of Monitoring and Measurement of Processes and product to ISO 13485? ISO 13485:2016 - Medical Device Quality Management Systems 1
Flexible "Ring" Roundness Measurement General Measurement Device and Calibration Topics 2
IATF 16949 SI # 10: Integrated self-calibration of measurement equipment- Needs for explanation IATF 16949 - Automotive Quality Systems Standard 2
QMS monitoring, measurement, analysis and evaluation requirement - Template ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 9
Physical Part Measurement General Measurement Device and Calibration Topics 6
Measurement Uncertainty in Optical Microscopy Measurement Uncertainty (MU) 1
ISO 6508 and portable hardness measurement instruments General Measurement Device and Calibration Topics 0
How to Establish the Calibration & Measurement Capability (CMC)? ISO 17025 related Discussions 1
Measurement of residual magnetism General Measurement Device and Calibration Topics 2
Surgical angle measurement guide device with an application software Medical Device and FDA Regulations and Standards News 4
Disabling measurement data during fault conditions IEC 60601 - Medical Electrical Equipment Safety Standards Series 5
Most versatile measurement device General Measurement Device and Calibration Topics 2
Surface scratch measurement General Measurement Device and Calibration Topics 1
Drawing tolerance vs. Measurement device General Measurement Device and Calibration Topics 7
MSA on automated measurement system - Multiple Step Vision System Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
Thermo-Hygrometer - Measurement of 10DegC @10%RH ISO 17025 related Discussions 4
GDT Flatness measurement question Inspection, Prints (Drawings), Testing, Sampling and Related Topics 10
Software Validation for Measurement Tools used in Process Validation ISO 13485:2016 - Medical Device Quality Management Systems 2
Best Measurement Systems Demos in California? General Measurement Device and Calibration Topics 1
Measurement Error - How to determine what is acceptable? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
Process Mapping - Inputs/Outputs/Detail Activities/Control points/Measurement? Process Maps, Process Mapping and Turtle Diagrams 4
IEC 60601-2-10 - Accuracy of Pulse Parameters - Required Measurement Uncertainty IEC 60601 - Medical Electrical Equipment Safety Standards Series 3
Program or application for standard time measurement in industry from video surveillance? Human Factors and Ergonomics in Engineering 1
Defining frequency of measurement tools callibration Calibration and Metrology Software and Hardware 3
How to “validate/qualify” a new measurement equipment? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 4