# Gage R&R with a unilateral specification tolerance - What about the other limit?

M

#### MANRAM

How can we calculate the R&R to a gauge if the process where is used only has unilateral tolerance? i.e. min 35 Lb-in

D

#### David Drue Stauffer

In the R&R process, the values are recorded and tabulated, the result is expressed as a percentage of the total tolerance. If the Low end of the tolerance is "0" and the high end of the tolerance is (-35) the absolute value of the total tolerance is "35". When the R&R Values are computed, squared and added together, then the square root is taken and divided by the tolerance, the resulting number is the gage R&R percentage.
10% is excellent, 11-30% is acceptable based upon the application, but should be closely monitored, and 31% or greater is unacceptable.

M

#### MANRAM

Understand what you say, and that work if you have only specified the limit at the high side of the spec, but what happen when you have only the low side of the spec.?

B

#### Batman

Perhaps you could use the actual error as a guide. When you calculate a GR&R, you get 2 values, the actual error and a percent of tolerance. With respect to your minimum requirement of 35lb-in, if the actual error in your R&R in this case is say 3lb-in, and you can demonstrate that your process is capable to +38lb-in minimum, then the actual gage error of 3 will not cause you to accept bad product. Sign it off and make parts.

S

#### simonchisholm55

Would this apply equally to leakage?
The maximum allowable is 4 ml min

Above 4ml min the part is rejected.

Attribute wise the gauge is capable, but our customer insists on
variable gauge R&R.

This is causing a headache, as my results are well over 30%

Simon

N

#### noboxwine

In my opinion, if the test for Product Conformity is Pass/Fail, regardless of registering a value, then an Attribute Study would apply. Your product doesn't perform any better at 41 than at 37, so you just need to hit that minimum target. That's why the specification is written as such. That is, only if the test doesn’t destruct the product or compromises the integrity. If that is the case, a Gage R&R couldn't appropriately be conducted, as the Reference Values could change during the study. In these cases, I have simply illustrated this to the customer and shown them records of calibrations to show that the load cell performs to the specification being used. Good Luck and let us know how you make out.

#### bobdoering

Trusted Information Resource
Would this apply equally to leakage?
The maximum allowable is 4 ml min

Above 4ml min the part is rejected.

This is causing a headache, as my results are well over 30%
What pressure is this test run at?

#### bobdoering

Trusted Information Resource
Test pressure is 1.38 bar +/- 60mb
I have been working on trying to get a good Gage R&R for a 5 cc/min at 3 bar, and it is tough. I get about .12 variation on just the zero. The total variation in a combination of normal and non-normal distributions. The non-normal distributions include the overall leak distribution, because the closer you get to zero, the less normal it is - conversely the further you get from zero the more normal it is. Similar to a Weibull curve. Then the individual measurement error is the same - you can get a good seat, or the best leak value, or any possibility of worse seats, fewer further away. Again, very similar to a Weibull distribution. Then, you get your zero error, which may be normal. Because these errors are significant a low leak rates, it is not easy to get a good gage R&R. At out 50 cc/min requirement, it was like a broad side of a barn with 3% GRR to tolerance.

Of course, customers do not realize this...

S

#### simonchisholm55

Thanks.

I will try again and use minitab to analyse the results. I agree however that as a Pass/Fail test
with calibration to +/- 0.05, then the equipment is biased towards failing good parts, NOT passing bad parts, therefore an attribute R&R is more appropriate.
Convincing our friends from OEM STA, is a harder nut to crack.

D Part Variation, Unilateral Tolerance, Destructive Testing - Gage R&R Experts pls help Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 12
J Unilateral Gage R&R (GR&R) - Minitab 13 - Does this study pass or fail and WHY? Using Minitab Software 17
T Gage R&R Using a Unilateral Tolerance - Only a minimum limit is specified Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
Gage calibration frequency, ISO and IATF - What are the requirements Calibration Frequency (Interval) 3
Determining Uncertainty from Gage R&R Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
Correct Calibration Method for Dial Depth Gage General Measurement Device and Calibration Topics 6
Gage Block Wringing General Measurement Device and Calibration Topics 3
Gage R&R with NDC=1 Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 19
Type 1 Gage R&R or something else? Reliability Analysis - Predictions, Testing and Standards 6
Should I perform Gage R&R only at the beginning of a new project? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 6
Thread Plug Gage Pitch GO Diameter out of spec AQL - Acceptable Quality Level 8
Gage R&R - Where am I going wrong? Part of a FAIR submission (Aerospace) Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
Gage R&R Excel templates Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
Gage R&R with more than 3 appraisers Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
Gage R&R Template AIAG 4th Edition IATF 16949 - Automotive Quality Systems Standard 3
Attributes SPC study - Attributive control (Go gage) Statistical Analysis Tools, Techniques and SPC 5
Capability or Gage R&R Study for Leak Tester? Reliability Analysis - Predictions, Testing and Standards 15
% Tolerance - Type 1 study on the gages, then a gage R&R (ANOVA) Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 4
Definitive answer on Type 1 vs Type 2 vs Type 3 Gage Study Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 0
Gage Management and Gage R&R Software General Measurement Device and Calibration Topics 1
Gage R&R Study on Load Cells - Large chemical blending tanks IATF 16949 - Automotive Quality Systems Standard 1
Minitab Type 1 Gage Study on True Position Question Measurement Uncertainty (MU) 1
Gage R&R on alloy Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 10
Need Help - Runout - Function Gage Inspection, Prints (Drawings), Testing, Sampling and Related Topics 7
Gage R&R studies for identical Devices - Need to confirm the requirement to perform them Reliability Analysis - Predictions, Testing and Standards 2
Stab Pin Size For Check Gage Inspection, Prints (Drawings), Testing, Sampling and Related Topics 3
Functional Gage for TOP applied at RFS Design and Development of Products and Processes 5
No reproducibility Error in Gage R&R? Reliability Analysis - Predictions, Testing and Standards 4
Template for Attribute Gage R & R wanted Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
Warning about Deltronic Gage Pins - The "Certification of Accuracy" document General Measurement Device and Calibration Topics 13
Gage type 1 study on CMM Capability, Accuracy and Stability - Processes, Machines, etc. 2
Gage R&R for Instant Measurement Machine Manufacturing and Related Processes 6
Gage R&R with large inter part variance - Torque Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 7
Do we need normal data for gage r&r studies? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
Do I need part variation while doing Destructive Variable Gage R&R MSA study Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 19
Gage R&R Standard Deviations Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 6
Informational Gage R&R: Swapping operators Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
Gage R & R for large amount of inspections IATF 16949 - Automotive Quality Systems Standard 10
Ring Gage Thermal Expansion General Measurement Device and Calibration Topics 4
Gage R&R for test precision or uncertainty prediction? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
Minor diameter of threaded ring gage General Measurement Device and Calibration Topics 1
Does Torque Gage Need Validation? Reliability Analysis - Predictions, Testing and Standards 4
Ignoring a Failed Gage R&R and upcoming external audit IATF 16949 - Automotive Quality Systems Standard 22
Scaling a Gage R&R - 2 holding fixtures each with 10 nests Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 7
Gage R&R Acceptable (10-30%), deduct Total Variation from Tolerance Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 12
Inexpensive Gage Blocks for small aerospace startup shop General Measurement Device and Calibration Topics 1
Position tolerance check gage manufacturing Inspection, Prints (Drawings), Testing, Sampling and Related Topics 8
Informational Expanded Gage R&R Analysis using Minitab Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 18
Interesting Discussion Threaded ring gage calibration and tendonitis in the elbows General Measurement Device and Calibration Topics 8
Federal 700B-26 Gage Block Comparator is not reading properly General Measurement Device and Calibration Topics 0