How can I measure three true positions? GD&T basic question

E

Emmamartsolf

I am more at electronics field. But now I have question about mechanical true position measurement. The drawing is attached. How can I measure three true positions?

Attachments

• 66.9 KB Views: 2,798

Michael_M

Trusted Information Resource
Re: GD&T basic question

In general (not taking into account max material (m)), measuring true position uses the following formula:

Determine the error (in X).....for example, print calls for a basic (boxed) dimension of 1.000 and you actually measure 1.0015, this gives us an error of .0015.
Determine the error (in Y).....for example, print calls for a basic dimension of .500 and you actually measure .4975, this gives us an error of .0025.

Take the error (in X---.0015) and square it .0015 x .0015=.00000225
Take the error (in Y----.0025) and square it .0025 x .0025=.00000625

Add the two together=.0000085 then take the square root of this--.002915476.

and finally, multiply the final result by 2 (The drawing is in Diameters, the formula above is used for radi, so we have to multiply by 2)

The final results must be less than or equal to the number in the box (.010 in your example).

You most likely will recognize this formula as the pythagorean theorem A squared + B squared = C squared

In your example, the holes on the outside are referenced to a 60 degree angle. Your example does not show the 'radius' to the hole or the bolt circle diameter but if you just calculate where the hole should be and use the same 'where it should be' vs. where it actually is (error) then it should work for you as well.

Hope this helps a little.

E

Emmamartsolf

Re: GD&T basic question

Thank you Michael. Appreciate your help.

The first TP Dia 0.395" -0.384", I think I can get center of Cir 1.812 (Plant A) to be (0,0), compare the center of this small Cir 0.390", use square root method to calculate the error, such as Squ((x-0.390)2+(y-0.390)2).

The second TP0.250 will be the same idea, with position of center line of the notch. It will be (w-0.250)

The third TP 0.148", is that the position of center of hole 0.148" ( M, N), then I use squ((M-0.148)2+(N-0.148))?

Thank you,

Emma

Michael_M

Trusted Information Resource
Re: GD&T basic question

It is hard to read the square and square root in a forum, but if I am reading this correctly, yes. Don't forget to multiply the final results by 2 after the square root is finished.

Michael_M

Trusted Information Resource
Re: GD&T basic question

I have had this PDF for a little while. It may help give a better understanding of Geometric tolerancing.

Attachments

• 380.5 KB Views: 905
E

Emmamartsolf

Re: GD&T basic question

Thank you. You are so helpful. Yes, actually, in equation, I mess radius and diameter. I should use radius, then double the error as the final result. But you are getting my thought.

Thank you for the attachment.

S Best wire size for 'three wire method' to measure pitch dia. of asymmetrical thread General Measurement Device and Calibration Topics 2
How to measure a plastic cap General Measurement Device and Calibration Topics 18
First Article Inspections - 'how to measure' General Measurement Device and Calibration Topics 11
Question regarding the calibration of monitoring and measure equipment. Oil and Gas Industry Standards and Regulations 0
Should a Class 1 medical device with an option to measure body weight be considered Class 1m? EU Medical Device Regulations 0
Gage R&R and right way to measure Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 16
How to measure DPPM in Chemical batch manufacturing Quality Tools, Improvement and Analysis 1
How to measure severity if my product is designed for emergency use and failure would result in death? ISO 14971 - Medical Device Risk Management 9
Looking for a third party to calibrate our measure equipment - South Carolina General Measurement Device and Calibration Topics 6
Validity / outcomes measure for custom made medical device ISO 13485:2016 - Medical Device Quality Management Systems 2
Performance specification as a Risk Control Measure, EN 14971 ISO 14971 - Medical Device Risk Management 7
KPIs or Metrics to Measure a New Complaint Handling Process 21 CFR Part 820 - US FDA Quality System Regulations (QSR) 9
Torque fractional unit of measure; cN-m or N-cm? General Measurement Device and Calibration Topics 2
ISO 9001 5.1.1 - How to measure System effectiveness ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 4
How to measure a Critical Characteristic that is a Basic Dimension Misc. Quality Assurance and Business Systems Related Topics 6
Can we use pin gauges to measure an accurate Cpk ? Capability, Accuracy and Stability - Processes, Machines, etc. 9
Best criteria to measure Corrective Action effectiveness - Poor Maintenance ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 10
1 How to measure straightness on a HEX Long Bar Manufacturing and Related Processes 7
Measurement Method Challenge - Measure feature #91 Inspection, Prints (Drawings), Testing, Sampling and Related Topics 9
Measure 3 times, for calibration report? Customer Owned Reference Standard General Measurement Device and Calibration Topics 4
Informational The ANSM decides, as a precautionary measure, to withdraw from the market macrotextured breast implants and breast implants with polyurethane-coated s Medical Device and FDA Regulations and Standards News 0
Informational Health Canada advises Allergan of its intent to suspend its licences for Biocell breast implants as a precautionary measure Medical Device and FDA Regulations and Standards News 0
U SI 13 - Not Every Process Requires an Efficiency Measure IATF 16949 - Automotive Quality Systems Standard 1
P NPR Metric - What if Customer has relaxed measure? TL 9000 Telecommunications Standard and QuEST 1
Tape Measure, the right device for 2 or 3 decimals accuracy on prints General Measurement Device and Calibration Topics 8
M How do we document and measure a service we do not provide? IATF 16949 - Automotive Quality Systems Standard 1
M For a Type 1 Gage R&R I want to use 1 unit and measure it 30 times Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
M How to Measure Paper considering Variation and Humidity General Measurement Device and Calibration Topics 10
Advice/recommendations for equipment to measure a soft pvc tube - Optical Measuring? General Measurement Device and Calibration Topics 4
R Height gauge to measure short hard to access heights General Measurement Device and Calibration Topics 3
D Proper Tool to Measure Wall Thickness (ISO 3611 - Micrometers) Manufacturing and Related Processes 7
S Quality Measure for Engineering Services Misc. Quality Assurance and Business Systems Related Topics 1
A What are methods and tools to measure Workload analysis ? Human Factors and Ergonomics in Engineering 1
M How do I measure variation in my process ? Capability, Accuracy and Stability - Processes, Machines, etc. 3
J How to measure process quality? Is DPPM the correct metric? Capability, Accuracy and Stability - Processes, Machines, etc. 5
M Micrometer Anvil Placement to measure the diameter of Steel Wires Measurement Uncertainty (MU) 6
S Where to Get SPC Training to Measure our Processes? Training - Internal, External, Online and Distance Learning 12
A question on Measurement System Analysis - Stability to measure got Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
Measure Twice, Cut Once: The French Railroad Snafu World News 18
S Measure water activity for fresh raw chicken meat and giblets Food Safety - ISO 22000, HACCP (21 CFR 120) 1
O Metrology Devices to Measure the Basic Dimensions of Plastic Specimens General Measurement Device and Calibration Topics 6
L How to Measure Paint Thickness for an Enclosure Manufacturing and Related Processes 5
C Chamfer Measurements - Do you measure the height and width or the hypotenuse? Manufacturing and Related Processes 15
Repeatability in MSA - Is repeatability a measure of Location Error Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
V How to use a Calibration Certificate in Uncertainty Calculation for a Test Measure Measurement Uncertainty (MU) 13
R How do companies measure QC Productivity? Quality Manager and Management Related Issues 3
V How to measure radius without radius gages APQP and PPAP 4
A Device to measure "Straightness & Deformation" of a metal that's under water Manufacturing and Related Processes 4
M How to measure effectiveness and efficiency of the established QMS as Auditors ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 14
F Specification for how to measure lengths of hose assemblies Various Other Specifications, Standards, and related Requirements 1