# Intro to Measurement System Analysis (MSA) of Continuous Data – Part 4: Stability

#### Miner

##### Forum Moderator
Staff member
This is the fourth in a series of articles about MSA. The focus of this article will be on measurement stability.

Stability is simply measurement bias throughout an extended period of time. This is where calibration falls short from an MSA perspective. Calibration is a series of snapshots widely spaced in time taken under controlled environmental conditions. A stability study is a series of repeated measurements taken under actual usage conditions. The purpose is to verify that the bias of the gage does not change over time due to environmental conditions or other causes.

A stability study is performed by selecting a measurement standard (ideal) or a master sample part that is midrange of the expected measurement range. Note: This may be enhanced by adding standards/master parts at the low and high ends of the expected measurement range. On a periodic basis, measure the standard 3 – 5 times. The period should be based on knowledge of what may influence the measurement system. For example, if ambient temperature variation is expected to be the major source of variation, make hourly checks throughout the day. If the source of variation is expected to be long term drift, take daily or weekly measurements.

Analyze the data using Xbar/R or Xbar/s control charts (use separate charts if you measured at the low/middle/high ends of the expected measurement range). The subgroups are comprised of the 3 -5 measurements and measure short term repeatability of the measurement device. If the control chart is in a state of statistical control throughout the study period, the gage stability is acceptable. There is no numerical acceptance criterion. If the control chart is out of control, analyze the patterns.
• For example, the influence of temperature would be expected to appear as cyclical trends that coincides with the ambient temperature.
• If the gage operates on plant utilities (e.g., air pressure) abrupt shifts could occur based on plant demand on the utilities (e.g., air pressure).
• Single points out of control could be the result of a gage that is overly sensitive to operator technique.
• Runs could be the result of different measurement methods
A stability study will also provide an estimate of the within operator repeatability of the gage. StdDevrepeatability = Rbar/d*2 (d2 may be used if the number of subgroups is greater than 20).

The next article will be:
Intro to Measurement System Analysis (MSA) of Continuous Data – Part 5: Repeatability & Reproducibility

Last edited by a moderator:

#### Marc

##### Hunkered Down for the Duration with a Mask on...
Staff member
Thank you!

Intro to Measurement System Analysis (MSA) of Continuous Data – Part 5b: R&R Imported Legacy Blogs 14
Intro to Measurement System Analysis (MSA) of Continuous Data – Part 5a: R&R Imported Legacy Blogs 5
Intro to Measurement System Analysis (MSA) of Continuous Data – Part 3: Linearity Imported Legacy Blogs 15
Intro to Measurement System Analysis (MSA) of Continuous Data – Part 2: Bias Imported Legacy Blogs 4
Intro to Measurement System Analysis (MSA) of Continuous Data – Part 1: What and Why? Imported Legacy Blogs 22
Intro to MSA of Continuous Data - Part 10: Creating gauge families Imported Legacy Blogs 10
Intro to MSA of Continuous Data - Part 9: %Tolerance for One-sided Specs Imported Legacy Blogs 3
Intro to MSA of Continuous Data – Part 8: Comparison of Two Gages Imported Legacy Blogs 3
Intro to MSA of Continuous Data – Part 7: R&R using Wheeler’s Honest Gage Study Imported Legacy Blogs 17
Intro to MSA of Continuous Data – Part 6: R&R for Non-Replicable Measurements Imported Legacy Blogs 18
T Intro & ISO 9001:2006 Question - Is ISO 9001:2006 Released? ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 1
E French Intro to Quality - Introducing ISO 9001 to New Employees Book, Video, Blog and Web Site Reviews and Recommendations 1
QMS monitoring, measurement, analysis and evaluation requirement - Template ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 3
Physical Part Measurement General Measurement Device and Calibration Topics 6
Measurement Uncertainty in Optical Microscopy Measurement Uncertainty (MU) 1
ISO 6508 and portable hardness measurement instruments General Measurement Device and Calibration Topics 0
How to Establish the Calibration & Measurement Capability (CMC)? ISO 17025 related Discussions 1
Measurement of residual magnetism General Measurement Device and Calibration Topics 2
Surgical angle measurement guide device with an application software Medical Device and FDA Regulations and Standards News 1
Disabling measurement data during fault conditions IEC 60601 - Medical Electrical Equipment Safety Standards Series 5
Most versatile measurement device General Measurement Device and Calibration Topics 2
Surface scratch measurement General Measurement Device and Calibration Topics 1
Drawing tolerance vs. Measurement device General Measurement Device and Calibration Topics 7
MSA on automated measurement system - Multiple Step Vision System Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
Thermo-Hygrometer - Measurement of 10DegC @10%RH ISO 17025 related Discussions 4
GDT Flatness measurement question Inspection, Prints (Drawings), Testing, Sampling and Related Topics 10
Software Validation for Measurement Tools used in Process Validation ISO 13485:2016 - Medical Device Quality Management Systems 2
Best Measurement Systems Demos in California? General Measurement Device and Calibration Topics 1
Measurement Error - How to determine what is acceptable? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
Process Mapping - Inputs/Outputs/Detail Activities/Control points/Measurement? Process Maps, Process Mapping and Turtle Diagrams 4
IEC 60601-2-10 - Accuracy of Pulse Parameters - Required Measurement Uncertainty IEC 60601 - Medical Electrical Equipment Safety Standards Series 3
Program or application for standard time measurement in industry from video surveillance? Human Factors and Ergonomics in Engineering 1
Defining frequency of measurement tools callibration Calibration and Metrology Software and Hardware 3
How to “validate/qualify” a new measurement equipment? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 4
Verification and Validation of Measurement Systems Misc. Quality Assurance and Business Systems Related Topics 0
How do you add accuracies for tools with different measurement devices? General Measurement Device and Calibration Topics 1
Viable count Measurement ISO Standard Other Medical Device Related Standards 1
Gauge R&R on a torque measurement Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 11
Air Quality Measurement Hardware and Software General Measurement Device and Calibration Topics 11
Dimension Measurement Tool Recommendation General Measurement Device and Calibration Topics 21
Does pitch/increment/resolution of a ruled scale apply to measurement uncertainty as line item? Measurement Uncertainty (MU) 10
Plane measurement with a 3D touch machine like a CMM. Calibration and Metrology Software and Hardware 3
Diameters measurement with 3D touch machine like a CMM Calibration and Metrology Software and Hardware 4
Uncertainty in measurement larger than tolerance Measurement Uncertainty (MU) 2
Measurement Uncertainties Budget for Thermometer Measurement Uncertainty (MU) 6
Performance Measurement ISO 9001: 2015 ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 2
Monitoring performance - How do I determine performance measurement basis within my organization? Misc. Quality Assurance and Business Systems Related Topics 4
What information to put on measurement Dimensional Results APQP and PPAP 7
O Ring capability and measurement - What is the automotive 'norm' for capability studies on O Rings? General Measurement Device and Calibration Topics 3
Electrical Metrology (Source/Generation and Measurement) General Measurement Device and Calibration Topics 1