MSA - How do I perform a Stability Study as a part of Measurement Systems Analysis?

N

NeilM

We are a company producing metal pressings and plastic injected moulded components to 2nd tier Automotive customers. As part of our QS9K audit in May I have been asked to implement stability studies alongside R&R studies, linearity studies (calibration) which are already in place.
I have started by taking three components produced in production, measuring each one ten times and recording the results (on a control plan feature). I have then calculated the sum, average, bias and range for each subgroup.
I have then plotted two graphs, the first being a linearity plot control chart & range chart. From this I have calculated the estimated Standard Deviation using a constant (d2) for sample size of 3. In the Ford MSA book it states that you must look for trends on the control chart as there is no need to calculate a measurement system stability number. Does anyone know if what I am doing will suffice ?? Your comments will be appreciated.

A

Al Dyer

Neilm,

It sounds similar to a method we have successfully implemented.

For a given fixture we have a calibrated master. Each Monday we take the master and verify (reset) the gage. We then measure the master five times and plot it on an X-R chart with a subgroup of five. Over time we have a picture of both the stability of the gage and also correlate it with any wear on the master, which is used to determine the calibration frequency of the master.

Short and sweet stability with added benefits.

ASD...

N

NeilM

My scenario is different as you have a calibrated master to work to and I only have a part that could measure differently depending on what method used

D

D.Scott

Without a calibrated master, the MSA recommends the following (paraphrased): use a production part that falls in the mid-range of the production requirements. Further, they recommend 3 samples from the low end, high end and mid-range and do seperate control charts for each. Then on a periodic basis (daily, weekly), measure the sample 3 to 5 times. Plot the data on an X-bar & R control chart. Establish your controls and evaluate for out of control or unstable conditions per the normal chart. Compute the S/D for the measurements and compare it with the process to determine if the system repeatability is suitable for the application. - Reading through your method I don't see measurements being taken over a period of time. I am a little confused when you add linerarity and bias to the stability study. The stability can stand on it's own while the others are best calculated on the 10 parts used in the actual R&R (just my opinion here - certainly not an expert on this).
Dave

N

NeilM

It is not possible for me to get 3 samples from the low end, high end and mid range of the tolerance as we produce these parts from a tool which has been cut to a fixed size. And therefore will measure what it measures (if thast makes sense!) Also is it important that we check parts so often as weekly?

N

NeilM

Also the parts we produce are often in metal which will only creep or shrink minimal amounts, which wouldnt even be picked up by the accuracy of normal measuring instruments. I.e vernier, caliper etc

D

D.Scott

You are losing me Neil - every process has variability and that variability can be measured if you use the right equipment. If your current gages don't pick it up, you are probably using the wrong gage. If the variance is minimal and insignifficant, why are you measuring at all?
You should be able to identify hi low mid parts just by isolating them when you find them through normal measurements.
Yes, the stability should be tracked at the outside weekly, although I don't know of any "rule" that says so. A longer time would introduce too many other possible variables to make me feel happy about it. We normally do daily stability checks on our Tensometer & Micro-Vu.
From what you have said, you may have to reconsider the measurement system you are using. Again - not my field so only an opinion.
Dave

MSA to perform Linearity and Stability studies on Variable Gauges with Minitab Ver 15 Using Minitab Software 6
How to perform a MSA - GRR (Gage R&R) A&R/ANOVA Vs GUM/VDA Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
T How to perform an MSA on Cross Hatch Test Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
G Tester Measurement Variability - How perform an MSA on a Test Machine Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
B How to perform MSA for PLC-interlocked torque tools Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
Torque wrenches and drivers - How to perform MSA (measurement system analysis) Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 9
N Do we have to perform MSA on every drawing dimension measured? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
M DMM Multimeter - Perform a complete MSA study? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 6
Updated MSA with the Annual PPAP validations APQP and PPAP 7
MSA for Push and Pull test Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
MSA for Calculated Clearance Dimension Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
MSA on automated measurement system - Multiple Step Vision System Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
MSA check list to audit IATF 16949 - Automotive Quality Systems Standard 8
MSA for ATE (Automatic Test Equipment Embedded Software) Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 9
MSA Tools Misc. Quality Assurance and Business Systems Related Topics 3
MSA studies for Heat Treatment Equipment Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
Industrial scales and MSA (IATF 16949 requirement 7.1.5.1.1) IATF 16949 - Automotive Quality Systems Standard 30
Difference between MSA and MSE? General Measurement Device and Calibration Topics 1
Test method validation - Is MSA (MSA1, MSA2, MSA3 and linearity) a good solution? Medical Device and FDA Regulations and Standards News 1
MSA results differences - Supplier results vs. My results Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
Preliminary MSA and Sockets' correlation Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 0
MSA requirement for 5 Micrometers + CP changes need customer approval? IATF 16949 - Automotive Quality Systems Standard 2
Correct MSA study for an automated camera system which makes attribute inspection Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
MSA Misc. Quality Assurance and Business Systems Related Topics 46
Measurement system analysis - Can you help me about implementation MSA for CMM device Manufacturing and Related Processes 2
MSA Study for a Leak Testing Device Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 6
Justifications for not performing MSA (Measurement System Analysis) Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 6
Do I need part variation while doing Destructive Variable Gage R&R MSA study Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 19
MSA Study - AS 9100 and and our customer want us to do MSA study for their parts dimension Reliability Analysis - Predictions, Testing and Standards 4
Compression Spring Force MSA Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
MSA Study on MTS dynamic rate measurements Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
The FILE recommended by AIAG MSA 4ed for LINEARITY Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 0
IATF 16949 Audit MSA IATF 16949 - Automotive Quality Systems Standard 10
Do Calibration Reference Standards require an MSA study? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
MSA When an Instrument Measures More than One Parameter Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 7
MSA Type 1 - Cg and Cgk ranges Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
MSA Study for a steel ruler Manufacturing and Related Processes 6
IATF 16949 MSA Studies for CMM machines IATF 16949 - Automotive Quality Systems Standard 3
7.1.5.1.1 MSA for inline dimension vision inspection equipment IATF 16949 - Automotive Quality Systems Standard 5
MSA on a Counting System Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
AIAG's IATF 16949 section 7.1.5.1.1 - Torque MSA Requirement Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 10
IATF 16949 7.1.5.1.1 Measurement System Analysis (MSA - Die casting) IATF 16949 - Automotive Quality Systems Standard 5
MSA on Part Specific Attribute Fixtures Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
ANOVA GR&R: Minitab vs AIAG MSA results Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 7
Test Method Validation or MSA for Batch Manufacturing Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
MSA on a process using different fixtures Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 1
B Camera Vision System Measurement System Analysis (MSA) Reliability Analysis - Predictions, Testing and Standards 3
D Cp/Cpk on Gages for MSA (vs. Cg/Cgk) Capability, Accuracy and Stability - Processes, Machines, etc. 21
Number of Distinct Categories" or NDC Calculation in MSA Studies Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 8
Q Doing an MSA on a Climatic Chamber Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3