# NIST HDBK 44 Table T.3. Class III Tolerance in Divisions Clarification

#### dgriffith

##### Quite Involved in Discussions
Table T.3. Class III Tolerance in Divisions (e), page 2-76, 2007 edition:
The first column header is TEST LOAD IN DIVISIONS CLASS III.
This is the applied mass load, but scaled in divisions instead of mass value?
Could someone help on this?

#### dwperron

Trusted Information Resource
Try checking Table 3, Note 4 on page 2-14 for an explanation:

The number of scale divisions, n, for each weighing range is determined by dividing the scale capacity for each range by the verification scale division, e, for each range.

#### dgriffith

##### Quite Involved in Discussions
Thanks for the response, and sorry, I should have been more clear.
I understand the divisions 'd' and 'e', but need to know what that first column header is referring to.
I interpret it to mean the test load is not in mass value (100 lbs, 50 kg, etc.), but in the number of divisions that mass load represents, which defines the tolerance in divisions, i.e. 500 lb scale @ 10,000 'e' divs yields a 0.05 division. A 500 lb test mass would represent 10,000 divisions, therefore the 4th line would indicate a tolerance of 5 divisions (0.25 lb) because the mass load is greater than 4001+ divisions.
Is that right or not, and if not, why.

#### dwperron

Trusted Information Resource
I believe you have it right.
That's the way I have interpreted the standard for determining tolerances on scales.

#### dgriffith

##### Quite Involved in Discussions
So, having established that the division "e" is 0.05 (500 lb FS and 10,000 division counts), if I then apply 150 lbs, do I do this?:
150 lb x 0.05 div/lb = 3000 div; the table 3rd line is 2000 - 4000 divisons, at that row the maintenance tolerance says now ?3 divisions.
Applying 150 lbs, the tolerance at that weight is ?3 div x 0.05 lbs/div = ?0.15 lbs.?
It seems to be a percentage of applied weight(reading).

Thread starter Similar threads Forum Replies Date
17025 and NIST handbook relationship (need advice) ISO 17025 related Discussions 8
Do these certificates of calibration meet ISO 9001 requirements for traceability to NIST? ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 12
Nist traceable calibration certificates General Measurement Device and Calibration Topics 1
What is the correct way to refer to NIST? Calibration Frequency (Interval) 4
NIST is one of the government services shut down General Measurement Device and Calibration Topics 2
NIST SOP 29 - Assignment of Uncertainty - Question General Measurement Device and Calibration Topics 0
S Reference calibration tools (traceable to NIST) that are not used in production ISO 13485:2016 - Medical Device Quality Management Systems 4
NIST’s Trove of Scientific Instruments and Relics After Work and Weekend Discussion Topics 2
Anyone working on NIST SP 800-171 (Network and Information Security)? Records and Data - Quality, Legal and Other Evidence 4
Traceability - Checklist from NIST General Measurement Device and Calibration Topics 4
ISO 13485 NIST Traceability requirements for calibration of test equipment ISO 13485:2016 - Medical Device Quality Management Systems 3
Is NIST traceability required if a Refractometer calibration is done in-house? Calibration Frequency (Interval) 6
C How to state that Calibration Services & Equipment are Traceable to NIST General Measurement Device and Calibration Topics 5
G Where to get a NIST Traceable Master Glass Refractive Index Sample General Measurement Device and Calibration Topics 5
A Calibration Service Suppliers refuse NIST reference numbers on C of C General Measurement Device and Calibration Topics 21
CMM Training at NIST - August 28-29, 2013 General Measurement Device and Calibration Topics 1
M NIST Standard for CO2 Monitor Calibration General Measurement Device and Calibration Topics 5
N NIST Special Publications and Technical Notes inquiry General Measurement Device and Calibration Topics 4
J Unbroken Chain to NIST or a Physical Property - Request for Peer Review General Measurement Device and Calibration Topics 6
S Does NIST Certify Calibration Laboratories? General Measurement Device and Calibration Topics 4
Outsourcing Calibration - NIST, OEM, and Calibration Laboratory General Measurement Device and Calibration Topics 2
S NIST equivalents for England and Germany Supplier Quality Assurance and other Supplier Issues 1
Does the NIST certificate that came with my new ball-bar have any value? General Measurement Device and Calibration Topics 8
Since NIST # can not be used to verify Traceability - What an Auditor should do AS9100, IAQG, NADCAP and Aerospace related Standards and Requirements 8
B NIST Traceable Thermometer Uncertainty Measurement Uncertainty (MU) 5
M Pipette Calibration & ISO 8655 - No NIST Standard(s) Traceability ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 5
New NIST reference materials available for popular diet supplement Pharmaceuticals (21 CFR Part 210, 21 CFR Part 211 and related Regulations) 0
S ISO 17025 requirements not enough to prove NIST Traceability? ISO 17025 related Discussions 10
NIST - US Revamps Science, Technology Standard-Setting Efforts General Measurement Device and Calibration Topics 0
NIST Traceability Numbers: Dense people, needle in a haystack General Measurement Device and Calibration Topics 16
R Does being traceable to NIST meet the requirements of ISO standards? General Measurement Device and Calibration Topics 16
N Thermocouple Drift Determination - Drift test done by NIST General Measurement Device and Calibration Topics 3
C Calibration Of Stop Watch using the NIST Clock General Measurement Device and Calibration Topics 4
C Steps to NIST - Measurement Traceability for a Certified Operator Class General Measurement Device and Calibration Topics 4
NIST Handbook of Mathematical Functions reference text World News 0
R NIST Test Number - What does it mean? General Measurement Device and Calibration Topics 3
Presentation About NIST Traceability General Measurement Device and Calibration Topics 8
J NIST Traceability: OEM Products - New electronics measurement device General Measurement Device and Calibration Topics 5
NIST comes out against DRE voting machines. Coffee Break and Water Cooler Discussions 7
Outside Calibration - Accredited, certified - Is NIST traceability guaranteed? General Measurement Device and Calibration Topics 8
Traceability - NIST Policy on Traceability, Section II.B.4 General Measurement Device and Calibration Topics 4
B Looking for Calibration / Reference Standards tracable to NIST General Measurement Device and Calibration Topics 15
R NIST - Primary Standard - Traceable to what? What exactly does that imply? General Measurement Device and Calibration Topics 10
Great NIST Stats Reference Statistical Analysis Tools, Techniques and SPC 2
S NIST Traceability - Calibrating a grid w/ laser interferonmeters and microscopes General Measurement Device and Calibration Topics 4
Uncertainty ListServe - NIST TN 1297 Measurement Uncertainty (MU) 3
M Calibration of Timers to NIST - How are digital and analog timers calibrated? General Measurement Device and Calibration Topics 15
NIST Traceability Numbers - Confused Yet? General Measurement Device and Calibration Topics 5
D Accreditation Scheme:relationships between ANSI, NIST, NCSL, ILAC, APLAC, NCCLS, etc. General Measurement Device and Calibration Topics 3
NIST 'High Integrity Software Systems Assurance' resource (site semi-abandoned) Software Quality Assurance 2