# Procedure for Comparing the Monitor Reading with Laboratory Result

L

#### lumpylumfy

I want to know what calculation this report used for calculated Bias formula.

Gas-in-oil standard S =100 ppm, laboratory result for S = 90 ppm
Laboratory result for 4 duplicate sample = 250, 230, 210 and 240 ppm
laboratory result corrected for bias = 275, 254, 231, 263 ppm

How they get (The red bold) result from bias calculation.

I just know its tell ' calculate the bias of the laboratory by comparing its results for the gas-in-oil standard sample to what was actually prepared'

I calculate the bias factor [10/11] mean 250 multiply with [10/11] get 274.

So what formula its use..here I attached the paper for reference.

I hope you all helping me out. I already google and find the answer for a month.

#### Attachments

• 113.3 KB Views: 176

#### Marc

##### Fully vaccinated are you?
My Thanks in advance to anyone who can help with this unanswered question. And, should you be browsing threads and find this one, and it still has no reply, if you can help we will all appreciate it.

#### Miner

##### Forum Moderator
I had no luck figuring out Annex E either.

L

#### lumpylumfy

I also read several journal and read again the report from CIGRE Report 409 Working Group D1.01 (TF 15) Report on Gas Monitor Oil-Filled Electrical Equipment . To looking again if I miss something.

Last edited by a moderator:

#### Statistical Steven

##### Statistician
Super Moderator
Looks like they correct the measured values by the percent bias between the standard and the lab result (100-90)/100 which is 10%. Multiple the unadjusted values by 10% you get the adjusted values.

I want to know what calculation this report used for calculated Bias formula.

Gas-in-oil standard S =100 ppm, laboratory result for S = 90 ppm
Laboratory result for 4 duplicate sample = 250, 230, 210 and 240 ppm
laboratory result corrected for bias = 275, 254, 231, 263 ppm

How they get (The red bold) result from bias calculation.

I just know its tell ' calculate the bias of the laboratory by comparing its results for the gas-in-oil standard sample to what was actually prepared'

I calculate the bias factor [10/11] mean 250 multiply with [10/11] get 274.

So what formula its use..here I attached the paper for reference.

I hope you all helping me out. I already google and find the answer for a month.

Bioburden requirements for Procedure Packs under Article 22? EU Medical Device Regulations 3
MD Design and development procedure ISO 13485:2016 - Medical Device Quality Management Systems 1
Biological evaluation procedure or plan EU Medical Device Regulations 3
Hi , everyone i need a procedure for validation of design prototype api 6d (valve manufacturing) Oil and Gas Industry Standards and Regulations 1
Does the MDR article 10 (14) needs to be documented word for word in a Procedure? EU Medical Device Regulations 1
HP 85054B Standard Mechanical Calibration Kit Calibration Procedure General Measurement Device and Calibration Topics 1
Procedure Pack SOP EU Medical Device Regulations 1
Inspection procedure for components Oil and Gas Industry Standards and Regulations 8
Writing Risk Management procedure for small manufacturing and we don't know where to start. Manufacturing and Related Processes 9
Bourdon gauge calibration procedure failure ISO 17025 related Discussions 3
Old procedure review and training requirements Document Control Systems, Procedures, Forms and Templates 9
Cease Production Sample Procedure Manufacturing and Related Processes 1
Risk Management Procedure updates needed for 14971:2019 ISO 14971 - Medical Device Risk Management 11
BASIC UDI-DI for a class IIa device marketed in a Procedure kit EU Medical Device Regulations 1
Statements into a procedure against audits ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 10
Internal and external communication procedure for Food Safety Food Safety - ISO 22000, HACCP (21 CFR 120) 2
Product Acceptance Software (PAS) PROCEDURE (BOEING D6-51991) AS9100, IAQG, NADCAP and Aerospace related Standards and Requirements 3
Design Control Procedure Medical Device and FDA Regulations and Standards News 4
Out of specification procedure for Medical device manufacturing Manufacturing and Related Processes 7
Risk Management File - Procedure Packs ISO 14971 - Medical Device Risk Management 3
Changes to procedure communication ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 7
R&D and Design procedures - can one procedure cover both activities Design and Development of Products and Processes 6
SPC procedure Statistical Analysis Tools, Techniques and SPC 5
Enough for PSUR to cover product family? (Procedure packs, not the legal manufacturer) EU Medical Device Regulations 1
Mock recall template/procedure ISO 13485:2016 - Medical Device Quality Management Systems 1
Help in document control procedure Document Control Systems, Procedures, Forms and Templates 14
Establishing a Data Analysis Procedure ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 6
Establishment and Product Registration Procedure - UK & EU UK Medical Device Regulations 0
Procedure for communication between authorized rep and distributor Other ISO and International Standards and European Regulations 0
Article 22 - Include statement in procedure pack? EU Medical Device Regulations 1
Classification of a Sterile Procedure Pack EU Medical Device Regulations 0
Management of change procedure for ISO 45001 Occupational Health & Safety Management Standards 5
Product and process Deviation procedure Manufacturing and Related Processes 1
IVD Metrological traceability without a reference measurement procedure General Measurement Device and Calibration Topics 11
GP-12 procedure form and requirements Customer and Company Specific Requirements 1
Procedure pack EU EU Medical Device Regulations 1
Process / Procedure - Radiographic (X-Ray) Non-Film Document Control Systems, Procedures, Forms and Templates 0
Procedure for vigilance system according to new MDR EU Medical Device Regulations 0
Procedure for clinical evaluation according to new MDR EU Medical Device Regulations 11
Production Work Order Procedure 21 CFR Part 820 - US FDA Quality System Regulations (QSR) 3
System&Procedure pack or accessory EU Medical Device Regulations 1
Do Contract Manufacturers need a complaint procedure? Medical Device and FDA Regulations and Standards News 10
Design and Development file Procedure ISO 13485:2016 - Medical Device Quality Management Systems 0
Procedure facility license in karnataka Manufacturing and Related Processes 3
MDR - System and procedure pack article 22 and all sub processes that apply ISO 13485:2016 - Medical Device Quality Management Systems 0
Software verification and validation procedure IEC 62304 - Medical Device Software Life Cycle Processes 6
Procedure packs with non-medical devices EU Medical Device Regulations 1
Need procedure for D&D inputs? ISO 13485:2016 - Medical Device Quality Management Systems 4
Example of a defined procedure for carrying out Material Review Board (MRB) Manufacturing and Related Processes 0
ISO 13485 procedure change and reflect to legacy manufacture items ISO 13485:2016 - Medical Device Quality Management Systems 2