Welcome to what was The Original Cayman Cove Forums! This thread is carried over and continued in the Current Elsmar Cove Forums

This is a "Frozen" Legacy Forum.
Discussions since 2001 are HERE

 The New Elsmar Cove Forums The New Elsmar Cove Forums   Measurement, Test and Calibration   MSA Manual    profile | register | preferences | faq | search
 Author Topic:   MSA Manual SamForum Contributor Posts: 244From:Registered: Sep 1999 posted 24 March 2000 12:29 PM                Having completed our QS9000 audit and now having idle time on my hands(sic)I have gone back to re-visit the MSA manual.Page 60 states that "If the analysis is based on tolerance . . . then substitute tolerance for total variation."I substituted the tolerance,.4mm, into the equation, the result 59% R&R. I would think the two results would have been closer. Are these typical results when using both options? What would I look for in the analysis to fix the problem?IP: Logged mbrunerForum Contributor Posts: 11From:Greenwood,IN,USARegistered: Feb 2000 posted 27 March 2000 12:13 PM                You didn't state what the %GRR results were based on total variation but I guess from the context that the total variation %GRR was less than the %GRR based on tolerance.First, regarding agreement between the two methods:They don't have to agree and they typically don't. Total variation is a combination of part variation & measurement system variation. Tolerance is the acceptable limits for the part value which may or may not have some relation to part variation. Generally you would hope that total variation is less than tolerance and that measurement system variation is much less than part variation.In this case %GRR based on tolerance would be less than %GRR based on total variation.In my experience at my company this is typical.What to look for in the analysis?Make sure you use the correct value for tolerance.If GRR% tolerance is greater than GRR% total variation the first item I check is the value for tolerance. GRR as percentage of tolerance uses the tolerance span. If the tolerance is +/- 0.4mm then the value that should be used in the calculation is 0.8mm not 0.4mm(0.8mm/sqrt(EV^2+AV^2).If the tolerance is a single sided spec, then the value for tolerance span is calculated by abs(2*(PartAverage-tolerance)).In this case the average measured value for the part influences the GRR result. The closer the part average is to the limit(tolerance) the smaller the tolerance span will be and thus makes tighter constraints on acceptable measurement system variation.If you can't reduce measurement system variation to an acceptable level in this case, check to see if the part average can be brought closer to nominal.I didn't mean to post such a long response. I don't profess to be an expert. I guess I'm just therapeutically bearing the scars of past mistakes.IP: Logged Marc SmithCheech Wizard Posts: 4119From:West Chester, OH, USARegistered: posted 27 March 2000 02:02 PM                Long, detailed responses are appreciated! The more folks help, the more these forums mean to each of us![This message has been edited by Marc Smith (edited 27 March 2000).]IP: Logged
 All times are Eastern Standard Time (USA) next newest topic | next oldest topic Administrative Options: Close Topic | Archive/Move | Delete Topic Hop to: Select a Forum or Archive List of Forums: Category: General Quality Topics --------------------Measurement, Test and CalibrationStatistical Techniques and 6 SigmaDocumentationNonconformance and Corrective Action SystemsContinuous ImprovementBenchmarkingTPM - Total Productive MaintenanceBaldrige National Quality AwardTQM - Total Quality ManagementMiscellaneous Quality TopicsBook ReviewsQuality Assurance SoftwareOccupations in Quality Assurance Category: National & International Standards --------------------ISO 9001/4:2000ISO 9000:1994ISO 14001 And Other Environmental SpecsGeneral ISO TopicsBS8800 & OHSAS 18001FDA and Medical DevicesAS 9100TL 9000 Category: Auditing and Registrars --------------------AuditingRegistrars and Registration Category: Automotive Related --------------------TS 16949QS-9000Tooling and Equipment SuppliersForo en Espa–olAPQP and PPAPFMEA and Control PlansEmbedded Devices and Control SoftwareVDA Volumes Including 6.1 Category: Guide Buyers Forums --------------------Guide Pre-Purchase QuestionsGuide Post-Purchase Questions Category: Software Quality Assurance --------------------Linux InfoTickit Scheme ForumSoftware Quality AssuranceInternet History Category: Odds and Ends --------------------Used - Buy and SellForums - How Do They Work? Suggestions!Company Advertisements - Free Offers, etc. List of Archives: PoliticsJokes