Welcome to what was The Original Cayman Cove Forums! This thread is carried over and continued in the Current Elsmar Cove Forums

This is a "Frozen" Legacy Forum.
Discussions since 2001 are HERE

 The New Elsmar Cove Forums The New Elsmar Cove Forums   Measurement, Test and Calibration   TAR vs TUR    profile | register | preferences | faq | search
 Author Topic:   TAR vs TUR ralutzLurker (<10 Posts) Posts: 1From:Gaithersburg, MD, USARegistered: Sep 2000 posted 26 September 2000 01:00 PM                Can anybody explain the differences between Test Accuracy Ratio (TAR) and Test Uncertainty Ratio (TUR)? IP: Logged awkForum Contributor Posts: 19From:Ontario, CanadaRegistered: Sep 2000 posted 26 September 2000 10:31 PM                ralutz:I am under the impression they are the same, used to evaluate and express measurement uncertainty, however I am not positive about this.The website for the National Institute of Standards and Technology, NIST, might provide you with a more informative answer.awkIP: Logged Jerry EldredForum Wizard Posts: 136From:Registered: Dec 1999 posted 27 September 2000 11:50 AM                They are pretty close, but I think there is a little difference between them.Uncertainty is the combined inaccuracy of a measurement. In a calibration context, if there are multiple instruments used to make a calibration measurement, if you used the Roo Sum Squares (RSS) method to calculate a total uncertainty, that would fall under the 'uncertainty' umbrella. The total uncertainty of a measurement may simply be the accuracy tolerance of a measurement standard. Or in the case of multiple instruments, and perhaps if environmental variation is a contributing factor, that amount of variation would also be factored in. Uncertainty refers to making a measurement.Accuracy on the other hand refers to the plus and minus tolerance of an instrument. TAR is a simpler quantity, referring to the ratios of the accuracies of two pieces of equipment. In calibration, uncertainty is prabably a more correct term. Where as in some other simpler applications, accuracy may be applicable. If you are using a hand held multimeter with a stated accuracy of +/- x% DC volts to adjust a power supply that has a stated accuracy of +/-y%, TAR would be appropriate. A complicated temperature calibration system with multiple instruments and opportunities for error would need to all be calculated together and a measurement uncertainty calculated.------------------IP: Logged Graeme C. Payneunregistered posted 29 September 2000 09:40 AM               Another way to approach this is to remember the subtle difference between accuracy and uncertainty.Accuracy is usually a design specification of a product feature. Uncertainty is always a performance characteristic of a measurement process.I usually explain the TAR/TUR difference like this:When you are using an instrument to measure a product, the specified accuracy (uncertainty) of the measuring instrument must be X times better than the accuracy tolerance of the feature being measured. That is a Test Accuracy Ratio.When you are calibrating a measuring instrument, the uncertainty performance of your calibration measurement process must be X times better than the measurement uncertainty (accuracy) of the instrument being calibrated. That is the Test Uncertainty Ratio.The difference is subtle, real, and usually does not make much difference. Graeme C. PayneASQ Certified Quality EngineerGraeme@asqnet.orgIP: Logged
 All times are Eastern Standard Time (USA) next newest topic | next oldest topic Administrative Options: Close Topic | Archive/Move | Delete Topic Hop to: Select a Forum or Archive List of Forums: Category: General Quality Topics --------------------Measurement, Test and CalibrationStatistical Techniques and 6 SigmaDocumentationNonconformance and Corrective Action SystemsContinuous ImprovementBenchmarkingTPM - Total Productive MaintenanceBaldrige National Quality AwardTQM - Total Quality ManagementMiscellaneous Quality TopicsBook ReviewsQuality Assurance SoftwareOccupations in Quality Assurance Category: National & International Standards --------------------ISO 9001/4:2000ISO 9000:1994ISO 14001 And Other Environmental SpecsGeneral ISO TopicsBS8800 & OHSAS 18001FDA and Medical DevicesAS 9100TL 9000 Category: Auditing and Registrars --------------------AuditingRegistrars and Registration Category: Automotive Related --------------------TS 16949QS-9000Tooling and Equipment SuppliersForo en Espa–olAPQP and PPAPFMEA and Control PlansEmbedded Devices and Control SoftwareVDA Volumes Including 6.1 Category: Guide Buyers Forums --------------------Guide Pre-Purchase QuestionsGuide Post-Purchase Questions Category: Software Quality Assurance --------------------Linux InfoTickit Scheme ForumSoftware Quality AssuranceInternet History Category: Odds and Ends --------------------Used - Buy and SellForums - How Do They Work? Suggestions!Company Advertisements - Free Offers, etc. List of Archives: PoliticsJokes