# Gage Blocks: Attribute or Variable Gage?

#### leaning

##### Involved In Discussions
Gage blocks:

(I posted this on ASQ's forum 2+ weeks ago. No answers yet, so I'm asking for EC's help.)

ASQ has them as "attribute gages".

QCI has them as "variable gages", but when I e-mailed them (QCI), one of the folks said that he calls them "variable" and the other calls them "attribute".

Anyone have any discussion on this so I can put this to bed?

Regards,
leaning

#### AMIT BALLAL

Trusted Information Resource
We can use it to check parameters such as slot and it can only tell whether it fits in or not (OK or Not OK). It can't tell us any reading by itself.

Therefore,
It is an attribute gage.

Last edited:

#### Ron Rompen

Trusted Information Resource
Although you CAN use them to make a variable measurement, gauge block stacks are generally considered to be attribute (go/no go) type gauges.

#### Bev D

##### Heretical Statistician
Staff member
Super Moderator
Well...it depends on what data results from the use of these gages. Are you generating pass/fail data or 'estimates' of a variables result in size? It can be either attributes or variables data...

Technically 'attribute' data is 'count' data that either has no natural rank or has a natural rank but violates other 'mathematical requirements' for adding, subtracting dividing and multiplying. These math requirements are:
• True zero
• Equal cell widths
• Equal distance between cell widths

So colors and gender have no natural rank, but categories of size, speed or other likely scale data (ordinal data) has rank but cannot be mathematically manipulated other than to describe the values in the categories as simple proportions of all of the categories. Pass/fail data falls into this category; it is nominal data. A simple count of values that fall into specified categories that are not equal in their bin size.

Attributes: Gage blocks and 'feeler gages' can be used to generate nominal data (pass/fail). Some dimension is either larger than or smaller than the gage block(s).

Variables: Gage blocks and feeler gages can also be used to determine a variables result. (Ratio data that meets all of the math requirements). Think about using a ruler tha only has 1/8" increments but your dimension is specified to 1/16". HOWEVER, this would be very chunky data; data with very little resolution. This type of data 'behaves' like ordinal data when we use statistical measures such as everage and standard deviation.

Inexpensive Gage Blocks for small aerospace startup shop General Measurement Device and Calibration Topics 1
Identification of gage blocks General Measurement Device and Calibration Topics 8
C How to Clean Gage Blocks and Gage Pins prior to Calibration General Measurement Device and Calibration Topics 4
S Purchase Order requirements Gage Blocks sent out for Calibration ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 1
Calibration Procedure for Gage Blocks General Measurement Device and Calibration Topics 10
V Excel Spreadsheet to Calculate Temperature Compensation of Gage Blocks wanted General Measurement Device and Calibration Topics 3
A Calibration Standards - Rod Standards vs. Gage Blocks Calibration Frequency (Interval) 2
P Gage Blocks Grades and Shapes. (What to purchase?) General Measurement Device and Calibration Topics 4
M Gage Blocks for a Mitutoyo Height Master 515-310 HM-12 General Measurement Device and Calibration Topics 1
In-House Calibration Requirements - Use of Grade 2 set of gage blocks General Measurement Device and Calibration Topics 2
C Gage R&R Study - Better to use parts or gage blocks? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 6
Need to find good set of Gage Blocks General Measurement Device and Calibration Topics 3
M What is the recommended calibration frequency for gage blocks? Calibration Frequency (Interval) 2
M Gage Block In-House Calibration - Blocks lost and users keep exchanging blocks General Measurement Device and Calibration Topics 2
M Calibrating Angle Gage Blocks General Measurement Device and Calibration Topics 2
Q Calibration through Gage Blocks/Test Fixtures/Customer Supplied ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 11
D Uncertainty Budget - Gage Blocks Measurement Uncertainty (MU) 7
C Gage Block Measurement - Home made gage blocks at 1" intervals IATF 16949 - Automotive Quality Systems Standard 7
Certifying Gage Blocks on an In-House CMM (Coordinate Measuring Machine) General Measurement Device and Calibration Topics 19
Gage R&R - Rockwell Scale - Hardness using Standard Test blocks Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 5
G Gage Calibration - Using a CMM and Gage Blocks for In-House Calibration ISO 9000, ISO 9001, and ISO 9004 Quality Management Systems Standards 1
G Serial Numbers on Gage Blocks? General Measurement Device and Calibration Topics 6
S Calibration of Micrometers - Can we put 2 gage blocks together? General Measurement Device and Calibration Topics 15
Gage Block Tolerances - Grade B Gage Blocks General Measurement Device and Calibration Topics 6
Gage Block Search - Looking for gage blocks that are 12, 24 and 40" General Measurement Device and Calibration Topics 5
B Need to know how to calibrate inch and metric gage blocks correctly. General Measurement Device and Calibration Topics 2
GGG-G-15 - Gage Blocks & Accessories - Cancelled July 1999 General Measurement Device and Calibration Topics 0
Gage Block Wringing General Measurement Device and Calibration Topics 3
Gage R&R with NDC=1 Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 19
Type 1 Gage R&R or something else? Reliability Analysis - Predictions, Testing and Standards 6
Should I perform Gage R&R only at the beginning of a new project? Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 6
Thread Plug Gage Pitch GO Diameter out of spec AQL - Acceptable Quality Level 8
Gage R&R - Where am I going wrong? Part of a FAIR submission (Aerospace) Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
Gage R&R Excel templates Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3
Gage R&R with more than 3 appraisers Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 2
Gage R&R Template AIAG 4th Edition IATF 16949 - Automotive Quality Systems Standard 3
Attributes SPC study - Attributive control (Go gage) Statistical Analysis Tools, Techniques and SPC 5
Capability or Gage R&R Study for Leak Tester? Reliability Analysis - Predictions, Testing and Standards 15
% Tolerance - Type 1 study on the gages, then a gage R&R (ANOVA) Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 4
Definitive answer on Type 1 vs Type 2 vs Type 3 Gage Study Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 0
Gage Management and Gage R&R Software General Measurement Device and Calibration Topics 1
Gage R&R Study on Load Cells - Large chemical blending tanks IATF 16949 - Automotive Quality Systems Standard 1
Minitab Type 1 Gage Study on True Position Question Measurement Uncertainty (MU) 1
Gage R&R on alloy Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 10
Need Help - Runout - Function Gage Inspection, Prints (Drawings), Testing, Sampling and Related Topics 7
Gage R&R studies for identical Devices - Need to confirm the requirement to perform them Reliability Analysis - Predictions, Testing and Standards 2
Stab Pin Size For Check Gage Inspection, Prints (Drawings), Testing, Sampling and Related Topics 3
Functional Gage for TOP applied at RFS Design and Development of Products and Processes 5
No reproducibility Error in Gage R&R? Reliability Analysis - Predictions, Testing and Standards 4
Template for Attribute Gage R & R wanted Gage R&R (GR&R) and MSA (Measurement Systems Analysis) 3